Descripción: manual electrocardiógrafo Cardiofax DECRETO LEGISLATIVO N° DEL REGIMEN DICIPLINARIO DE LA PNP (RESUMIDO). docx. D.L. DECRETO LEGISLATIVO QUE REGULA EL REGÍMEN DISCIPLINARIO DE LA PNP TITULO PRELIMINAR PRINCIPIOS. Grupo Ascenso PNP. LEY DEL RÉGIMEN DISCIPLINARIO DE LA PNP – DECRETO LEGISLATIVO Nº DEL 11 DE DICIEMBRE DE
|Published (Last):||1 April 2010|
|PDF File Size:||9.18 Mb|
|ePub File Size:||3.30 Mb|
|Price:||Free* [*Free Regsitration Required]|
Manual Cardiofax Home Manual Cardiofax Bmw Rr Motorcycle Service Manual. Cardiofax Gem Ecgk k. Marenbonearly Medieval Philosophy legizlativo Introduction. Hace un recuento de todas las funciones de un supervisor de barrido para el censo efextado por el DNP. Keep this manual near the instrument or in the reach of the operator and refer to it whenever the operation is unclear.
This product stores personal patient information. Manage and operate the information appropriately. Patient names on the screen shots and recording examples in this manual are fictional and any resemblance to any person living or dead is legiwlativo coincidental. The contents of this manual are subject to change without notice. Copyright Notice The entire contents of this manual are copyrighted by Nihon Kohden.
All rights are reserved. No part of this document may be reproduced, stored, or transmitted in any form or by any means electronic, mechanical, photocopied, recorded, or otherwise without the prior written permission of Nihon Kohden.
Trademark The mark printed on the SD card that is used in this instrument is a trademark. The company name and model name are trademarks and registered trademarks of each company. Use only Nihon Kohden approved products with this device.
EQWEQ by Siete Tacsi Avila on Prezi
Use of non-approved products or in a non-approved manner may affect the performance specifications of the device. This includes, but is not limited to, batteries, recording paper, pens, extension cables, electrode leads, input boxes and AC power.
Please read these precautions thoroughly before attempting to operate the instrument. To safely and effectively use the instrument, its operation must be fully understood.
When installing or storing the instrument, take the following precautions: Avoid vibration and mechanical shock, even during transport.
Before Operation 1 Check that the instrument is in perfect operating order. During Operation 1 Both the instrument and the patient must receive continual, careful attention. To Shutdown After Use 1 Turn power off with all controls returned to their original positions. The instrument must receive expert, professional attention for maintenance and repairs.
When the instrument is not functioning properly, it should be clearly marked to avoid operation while it is out of order. The instrument must not be altered or modified in any way. Maintenance and Inspection 1 The instrument and parts must undergo regular maintenance inspection at least every 6 months.
When the instrument is used with a defibrillator, make sure that the instrument is protected against defibrillator discharge. However, consumable materials such as recording paper, ink, stylus and battery are excluded from the warranty. NKC will not recognize any other warranty, either implied or in writing. In addition, service, technical modification or any other product change performed by someone other than NKC or its authorized agents without prior consent of NKC may be cause for voiding this warranty.
Defective products or parts must be returned to NKC or its authorized agents, along with an explanation of the failure. Shipping costs must be pre-paid. This warranty does not apply to products that have been modified, disassembled, reinstalled or repaired without Nihon Kohden approval or which have been subjected to neglect or accident, damage due to accident, fire, lightning, vandalism, water or other casualty, improper installation or application, or on which the original identification marks have been removed.
The following describes some common interference sources and remedial actions: Strong electromagnetic interference from a nearby emitter source such as an decretto radio station or cellular phone: Identify the cause of this interference and if possible remove this interference source. If this is not possible, use a different power supply.
Effect of direct or indirect electrostatic discharge: A humid room can help lessen this problem. Electromagnetic interference with any radio wave receiver such as radio or television: Use with other equipment: Use of unspecified configuration: Measurement with excessive sensitivity: When unexpected artifact appears, inspect the surrounding electromagnetic conditions and remove this artifact source. If the above suggested remedial actions do not solve the problem, consult your Nihon Kohden representative for additional suggestions.
The CE mark is a protected conformity mark of the European Community. Contact your Nihon Kohden representative for disposal. CAUTION A caution alerts the user to possible injury or problems with the instrument associated with its use or misuse such as instrument malfunction, instrument failure, damage to the instrument, or damage to other property.
NOTE A note provides specific information, in the form of recommendations, prerequirements, alternative methods or supplemental information.
For Nihon Kohden products marked with this symbol, contact your Nihon Kohden representative for disposal.
Failure to follow this warning may cause explosion or fire. Never use the electrocardiograph in a hyperbaric oxygen chamber.
Manual Cardiofax 1150
Otherwise, the current from the ESU flows into the electrodes of the electrocardiograph, causing electrical burn where the electrodes are attached. For details, refer to the ESU manual. When performing MRI test, remove all electrodes and transducers from the patient which are connected to this instrument.
Failure to follow this warning may cause skin burn on the patient. Lgislativo details, refer to the MRI manual. Before defibrillation, all persons must keep clear of the bed and must not touch the patient or any equipment or cable connected to the patient. Failure to follow this warning may cause electrical shock or injury. Failure to follow this warning may cause skin burn where the electrode is attached and damage the electrocardiograph due to discharge energy when defibrillation is performed.
Failure to follow this warning may result in electrical shock or injury to the patient and operator, and cause fire or instrument malfunction.
When the provided power cord cannot be used or when equipotential grounding is doubtful such as in a facility with poor groundingoperate the electrocardiograph on battery power. Otherwise, the patient and operator may receive electrical shock or injury.
Any potential difference between instruments may cause electrical shock to the patient and operator. Never use a local purchase multiple portable socket outlet. Failure to follow this warning may cause electrical shock to the patient and operator. If the multiple portable legiskativo outlet gets dusty, damaged or legislatvio liquid, it may cause instrument malfunction and electrical shock to the patient or legislattivo. To use non-medical equipment that requires an ddecreto transformer, only connect it 11150 a specified multiple portable socket outlet that has an isolation transformer.
Do not connect the nonmedical equipment to a wall AC socket or to a multiple portable socket that does not have an isolation transformer. Do not connect unspecified electrical instruments to a multiple portable socket outlet in the system.
Failure to follow this warning may cause leakage current to increase and the patient and operator may receive pnpp shock or injury. The impedance of the protective earth increases and the patient and operator may receive electrical shock or injury. If another cart is used, it may tip over or the electrocardiograph may fall off. The cart may tip over. To prevent the cart from tipping over or the electrocardiograph falling off the cart: Do not lean on the handle or put your weight on the cart.
If they are swallowed, consult a physician immediately. Using unspecified recording ppnp or electrodes may cause incorrect ECG recording and accelerate deterioration of the electrocardiograph. It may cause leakage, overheating, explosion and fire. If the liquid inside the battery contacts the eyes, mouth or skin, do not rub the eyes, mouth or skin. Wash with clean water and see a physician immediately. Do not charge a deteriorated battery pack.
Charging a deteriorated battery pack may cause the electrocardiograph to fail to deecreto in both battery and AC power. It may cause leakage of the battery pack and degrade the performance and lifetime of the battery pack. Before disposing of the battery pack, check with your local solid waste officials for details in your area for recycling options or proper disposal.
The battery is recyclable. At the end of its useful life, under various state and local laws, it may be illegal to dispose of this battery into the municipal waste stream. Failure to follow this warning may cause electrical shock or injury to the patient. Do not decrfto the tip of the electrode lead to anything other than an electrode.
Connect all the tips to the corresponding electrodes and attach them to the patient. If unconnected tips or electrodes contact an item which is not equipotential grounded, the patient may receive electrical shock. Also, noise may superimpose on the ECG waveforms and measurement data may be incorrect.
When managing examination data that is only in electronic file, make sure to enter the patient information. Do not use the output signal from the output connector of the electrocardiograph for a synchronization signal on a defibrillator.
There is a time delay between the input signal and output signal. When using the output signal from the electrocardiograph for the synchronization signal on other instrument, always consider this time delay.